Queensgate QI DPT-C-L Digital Piezo Translator & Control Module CM-LN-LD (3681)
Queensgate QI DPT-C-L Digital Piezo Translator and Control Module CM-LN-LD (MAS)
Control Module Low Noise and Low Drift
This unit was set at factory as "MASTER." It appears that you can go in and change this setting if needed. See instructions below that were copied from manufacturer's site.
Control Module Synchronisation When one or more Control Modules are operated from the same power supply, then noise may be introduced from one module to the other. This is due to the 30kHz oscillators in each module having slightly different frequencies causing "beating" effects. This can be eliminated by using one Control Module (MASTER) producing the 30kHz reference signal and the other modules (SLAVES), which have their oscillators disabled, use this reference signal. When the Control Modules are set up, pin 10 of the 37-way D-type connector is used to distribute the reference signal from the MASTER module to the SLAVE module(s). 4.1 Master Module Configuration a) Disconnect power cable from the Control Module. b) Remove CM lid (by removing 4 screws - 2 on each side). c) Move link from LNK11B (SLAVE) to LNK11A (MASTER) on "bottom or main" PCB - see figure 19. Note: LNK11A - MASTER is populated at the factory unless specifically set as "SLAVE" module. d) On "TOP" PCB, move LNK1 from "STD" to "SYNC". e) Replace the CM lid. 4.2 Slave Module Configuration a) Disconnect power cable from the Control Module. b) Remove CM lid (by removing 4 screws - 2 on each side). c) Move link from LNK11A (MASTER) to LNK11B (SLAVE) on "bottom or main" PCB - see figure 19. d) On "TOP" PCB, move LNK1 from "STD" to "SYNC" - see figure 19. e) Replace the CM lid.
Sold as is with 30 day Right of Return, only pictured items included
Please call 805.680.7222 for easy return/refund for any reason
Anacapa Equipment strives to offer excellent customer service. You may return any equipment purchased from us for any reason within thirty (30) days after your receipt of the equipment for a full refund of the purchase price, provided that returned items must be received in the same condition as sent.
All equipment purchased is shipped within one (1) day (subject to freight availability) after receipt of payment and is shipped FOB place of shipment. We only charge actual shipping cost. If eBay overcharges for shipping, we issue an immediate refund for the difference. If you are undercharged, we will absorb the cost.
Buyer acknowledges that the equipment is being sold “as is” and without any warranty of merchantability or fitness for a particular use. Buyer may inspect the equipment prior to purchase and placement of the order prior to inspection is deemed a waiver of Buyer’s right to inspect. Buyer is responsible for determining the suitability of this product for its application. Anacapa Equipment Brokers, Inc. assumes no liability for the use or misuse of any equipment purchased by Buyer. It is assumed that the unit will be installed and operated by qualified personnel. The purchaser agrees that the seller shall not be held responsible or liable for any injuries or damages, whether incidental or consequential, associated in any way with the equipment purchased. The Buyer, by purchasing this equipment, indicates its acknowledgment of, and agreement, to the terms of this limited warranty.
This equipment is not intended to be used, and should not be used in human experimentation, nor should this instrument be applied to humans in any way.